Note: You are seeing this text-only version of our site
because you are using an older browser that is not standards-compliant.
How to upgrade to a FREE standards-compliant browser.
Some Netscape 7 and Mozilla users may see this warning in error.
to fix the problem.
Back to Company
Continuously-flowing sample enters the turbidimeter and flows through a patented bubble removal system, which vents entrained air from the sample stream.
Meets and exceeds USEPA Method 180.1 for measuring turbidity for drinking water compliance; One controller accepts up to two plug-and-play sensors giving you two complete turbidimeters; Sensitive to fine changes in low level turbidity with a built-in bubble removal system; Built-in datalogger stores up to 6 months of data; Versatile digital output-Modbus protocol, LonWorks or wireless IR port
Range: 0 to 100 nephelometric turbidity units (NTU); Accuracy (Defined According to ISO 15389): ? 2% of reading or ? 0.015 NTU (whichever is greater) from 0 to 40 NTU; ? 5% of reading from 40 to 100 NTU; ? 5% of reading from 40 to 100 NTU
Copyright © 2005-2018 Modbus Organization, Inc.
PO Box 628 Hopkinton, MA 01748. All rights reserved.
By using Modbus.org, you accept the terms of our visitor